Infant Radiant Warmer with X-ray cassette
(Baby Radiant Warmer)
General of the Baby Radiant Warmer:
Three control modes: Pre-warm mode, manual mode and
baby mode controlled by micro-computer;
Set temperature and baby temperature can be displayed separately.
Tri-safety over-temp protection:
Double CPU control system.
Isolated over heating cutoff device.
Horizon angle of heater head and inclination of the bassinet can be
The glass panels around bassinet can be disassembled or turned
With self-checking function, various fault failure alarms.
X-ray cassette under the bassinet.
With temperature data storage function.
Standard configuration of the Infant Radiant Warmer:
Main body includes the warmer module, bassinet, main column, stand,
controller, I.V. Pole and tray.
Specification of the Baby Radiant Warmer:
Power requirement: AC220V-230V/50Hz, 600W
Mode of control: Pre-warm mode, manual mode and baby mode
controlled by micro-computer.
Baby mode temperature control range:34.5~ 37.5℃
Temperature sensor display range: 5℃~65℃
Accuracy of skin temperature sensor: ≤0.3℃
Temperature uniformity of mattress: ≤2℃
Angle of warmer module: 0°30°60°for two ways
Inclination of the bassinet: the three tile position.
APGAR timer: it will give sound when the unit tones at 1’,5’,10’
Failure alarm: over temperature alarm, deviation alarm, sensor
failure alarm, power failure alarm, setting alarm, checking alarm
and so on.
Operation range: 18℃~30℃
Ambient air movement rate: ≤0.3m/s
Package of the Baby Radiant Warmer:
Package: each unit is packed to one carton
Size: (Main body) 122cm x 67cm x 110cm.Gross Weight: 100kg